I just got my first implant (a NExT) a few days ago and while the high frequency side works nicely, I am having problems with the low frequency.
I have a Proxmark3 Easy, and after reading a few posts of the forum I was able to find a position where I could get consistent reading of the EM410 default of the T5577. However, when I tried a “t55xx detect” I couldn’t get a good response, and it always returned the message:
[!] Could not detect modulation automatically. Try setting it manually with 'lf t55xx config'
Note that I did configure the proxmark with the correct settings.
After a few search in proxmark3 forums, I tried to do a “t55xx wipe”, but since then the implant seems to be unresponsive to any further commands. I cannot read or detect anything, and after reading some scary posts on the tear protection I am afraid that the implant may be bricked.
Does anyone have any advice on how to proceed from here? Any ideas I can try?